Distance Measurements with Lasers
|<<< | >>> | Feedback|
Your are not yet listed? Get your entry!
Ask RP Photonics for advice concerning devices for distance measurements, such as interferometers.
German: Distanzmessung mit Lasern, Laser-Entfernungsmessung
Lasers can be used in various ways to measure distances or displacements without physical contact. In fact they allow for the most sensitive and precise length measurements, for extremely fast recordings (sometimes with a bandwidth of many megahertz), and for the largest measurement ranges, even though these qualities are usually not combined by a single technique. Depending on the specific demands, very different technical approaches can be appropriate. They find a wide range of applications, for example in architecture, inspection of fabrication halls, criminal scene investigation (CSI), and in the military.
Techniques for Distance Measurements
Some of the most important techniques used for laser distance meters are as follows:
- Triangulation is a geometric method, useful for distances in the range of ≈ 1 mm to many kilometers.
- Time-of-flight measurements (or pulse measurements) are based on measuring the time of flight of a laser pulse from the measurement device to some target and back again. Such methods are typically used for large distances such as hundreds of meters or many kilometers. Using advanced techniques, it is possible to measure the distance between Earth and the Moon with an accuracy of a few centimeters. Typical accuracies of simple devices for short distances are a few millimeters or centimeters.
- The phase shift method uses an intensity-modulated laser beam.
Compared with interferometric techniques, its accuracy is lower, but it allows unambiguous measurements over larger distances and is more suitable for targets with diffuse reflection.
Note that the phase shift technique is sometimes also called a time-of-flight technique, as the phase shift is proportional to the time of flight, but the term is more suitable for methods as described above where the time of flight of a light pulse is measured.
- For small distances, one sometimes uses ultrasonic time-of-flight methods, and the device may contain a laser pointer just for getting the right direction, but not for the distance measurement itself.
- Frequency modulation methods involve frequency-modulated laser beams, for example with a repetitive linear frequency ramp. The distance to be measured can be translated into a frequency offset, which may be measured via a beat note of the sent-out and received beam.
- Interferometers allow for distance measurements with an accuracy which is far better than the wavelength of the light used.
A laser radar is a device which uses one of the distance measurement techniques as described above, and scans the direction of the distance measurement in two dimensions. This allows the acquisition of an image, or more precisely a depth profile of some object, as required e.g. in robotics. For acquiring such depth profiles at a higher rate, there are sensor chips similar to CCDs (charge-coupled devices) with internal electronics to detect phase shifts, so that the distance for each pixel can be measured simultaneously. This allows for rapid three-dimensional imaging with very compact devices.
Compared with ultrasonic or radio and microwave frequency devices (radar), the main advantage of laser distance measurement techniques is that laser light has a much smaller wavelength, allowing one to send out a much more concentrated probe beam and thus to achieve a higher transverse spatial resolution. Another advantage that an optical bandpass filter makes it possible to very effectively remove noise influences at other optical frequencies.
As essentially all other measurement techniques using lasers, laser distance measurements can be affected by laser noise. Other noise-related issues can arise from detection noise, stray light, and speckle effects.
The targets can have very different reflection and scattering properties. Problems can arise for very low reflection or for specular reflections.
Note that range finding with lasers can raise serious laser safety issues, particularly when intense pulses from Q-switched lasers are used. The related hazards can be strongly reduced by applying eye-safe lasers.
|||H. Kikuta et al., “Distance measurement by the wavelength shift of laser diode light”, Appl. Opt. 25 (17), 2976 (1986)|
|||G. Beheim and K. Fritsch, “Range finding using frequency-modulated laser diode”, Appl. Opt. 25 (9), 1439 (1986)|
|||T. Bosch et al., “The physical principles of wavelength-shift interferometric laser rangefinders”, J. Opt. 23, 117 (1992)|
|||C.-M. Wu et al., “Heterodyne interferometer with subatomic periodic nonlinearity”, Appl. Opt. 38 (19), 4089 (1999)|
|||M.-C. Amann et al., “Laser ranging: a critical review of usual techniques for distance measurement”, Opt. Eng. 40 (1), 10 (2001)|
|||S. Poujouly and B. Journet, “Laser range-finding by phase-shift measurement: moving toward smart systems”, K. G. Harding, J. W. V. Miller, and B. G. Batchlor, eds., Machine Vision and Three-Dimensional Imaging Systems for Inspection and Metrology, Proc. SPIE 4189, 152, SPIE (2001)|
|||T. R. Schibli et al., “Displacement metrology with sub-pm resolution in air based on a fs-comb wavelength synthesizer”, Opt. Express 14 (13), 5984 (2006)|
|||K. Joo et al., “Distance measurements by combined method based on a femtosecond pulse laser”, Opt. Express 16 (24), 19799 (2008)|
|||I. Coddington et al., “Rapid and precise absolute distance measurements at long range”, Nature Photon. 3, 351 (2009)|
|||G. Berkovic and E. Shafir, “Optical methods for distance and displacement measurements”, Adv. in Opt. and Photon. 4 (4), 441 (2012)|