Infrared detectors are detectors for infrared light. Some of them are photodetectors (photon detectors) which are sensitive in a relatively long wavelength region, while others are based on sensing small temperature increases induced by absorption of infrared light.
One may in principle also regard laser viewing cards as infrared detectors, which however do not deliver a signal, but emit some visible light.
For the short-wavelength part of the near infrared, ordinary silicon photodiodes can be used; they are responsivity falls off for wavelengths somewhere between 1 μm and 1.1 μm, depending on the thickness of the active region.
For longer wavelengths, semiconductor materials with a lower band gap energy are required, so that photons with lower energy are sufficient to create a carrier which contribute to some photocurrent. Indium gallium arsenide (InGaAs) detectors are suitable for wavelengths up to ≈1.7 μm. Similarly, germanium (Ge) photodiodes can be used at ≈0.9 μm to 1.6 μm. For longer wavelengths even beyond 5 μm, indium antimonide (InSb) photodiodes are available.
Infrared Photoconductive Detectors
Infrared light at wavelengths sometimes beyond 5 μm can be detected with lead salt detectors, e.g. containing lead selenide (PbSe) or lead sulfide (PbS). Such devices are not fabricated with epitaxial growth, as e.g. InGaAs photodiodes, but rather by deposition of thin polycrystalline films. They are photoconductors (photoresistors), not photodiodes, i.e., they do not contain a p–n junction but exhibit a reduction in electrical resistance caused by incident light, which induces intraband transitions. Lead salt detectors can be used at room temperature, although better performance is possible with cooled detectors. Note that cooling does not only reduce the dark current and therefore improve the sensitivity, but also reduces the band gap energy, so that light with longer wavelengths can be detected.
Another detector technology, which is widely used, is based on photoresistors containing mercury cadmium telluride (MCT, HgxCd1−xTe). Through the composition parameter (i.e., the mercury content), the bandgap energy can be tuned in a wide range. That way, detection at rather long wavelengths (partly beyond 12 μm) is possible. MCT detectors are usually cooled, and they can be made as image sensors, which are mostly used for thermal imaging.
For even longer infrared wavelengths – even in the far infrared somewhat beyond 200 μm wavelength –, there are highly doped silicon and germanium detectors. Such long-wave infrared detectors are extrinsic (impurity band) semiconductor devices, i.e., they rely on additional energy levels introduced by impurities like arsenic, copper, gold or indium. They need to be cooled to very low temperatures (e.g. 4 K with liquid helium), because thermal excitation would otherwise mask any infrared light signals.
Another technology is that of quantum well infrared photodetectors (QWIP) based on GaAs/AlGaAs. They are cooled photoconductive detectors containing multiple quantum wells. At low operation temperatures of e.g. 50 K, they have some advantages over MCT detectors, particularly for the fabrication of imaging detector arrays. They can work up to wavelengths of ≈19 μm.
Some of the infrared detector materials contain hazardous substances like lead, cadmium and mercury. Therefore, their use is increasingly restricted, e.g. by the European RoHS regulations. Temporary exceptions, however, are in place for specific applications, where it is difficult to find substitutes.
Thermal Infrared Detectors
- There are pyroelectric materials such as barium strontium titanate, generating a small electric voltage in response to a temperature gradient. The generated heat is conducted to some substrate, and the temperature gradient is proportional to the incident intensity.
- There are bolometers, which can also be made in the form of microbolometers. Such a sensor contains a thin infrared-absorbing plate, made e.g. of amorphous silicon (aSi) or vanadium oxide, which is suspended with two electrodes. One can then measure the temperature-dependent electrical resistance of the plate. Note that without incoming infrared light the temperature of the plate will relax towards the substrate temperature, while incoming radiation will increase the temperature above that of the substrate. The readings will thus have to be corrected according to the substrate temperature, if that is not held constant. A metal mirror below the plate may be used to back-reflect non-absorbed infrared light and at the same time to reduce the thermal radiation from the substrate.
Thermal detectors are often operated at room temperature, i.e., without cooling. Often, however, they are temperature-stabilized, i.e., the operation temperature is kept constant, typically with a small Peltier cooler, but not at a particularly low temperature. With cooled operation, one can further improve the performance.
If you like this article, share it with your friends and colleagues, e.g. via social media: