Time-of-flight measurements are often used for the measurement of some distance, e.g. with a laser rangefinder, used e.g. in an airplane, possibly in the form of a scanning laser radar. Here, an apparatus sends out a short light pulse and measures the time until a reflected portion of the pulse is monitored. The distance is then calculated as half the measured round-trip time divided by the velocity of light. Due to that high velocity, the temporal accuracy must be very high – e.g. 1 ns for a spatial accuracy of 15 cm.
The time-of-flight method is typically used for large distances such as hundreds of meters or many kilometers. Using advanced techniques (involving high-quality telescopes, highly sensitive photodetection, etc.), it is possible to measure e.g. the distance between the Earth and the Moon with an accuracy of a few centimeters, or to obtain a precise profile of a dam. Typical accuracies of simple devices for short distances are a few millimeters or centimeters.
For a much higher spatial resolution, ultrashort pulses can be used, e.g. from a femtosecond laser. However, fast photodetectors (usually photodiodes) are limited in resolution at least to the picosecond region. Femtosecond temporal resolutions require all-optical techniques such as measuring optical cross-correlations.
As time-of-flight measurements are preferentially used for large distances, the beam quality of the laser source is crucial. In addition, a telescope can be used to obtain a large beam diameter and an accordingly increased Rayleigh length, i.e. a small beam divergence. The target may be equipped with a retroreflector in order to increase the amount of reflected light. The pulse duration used is usually between 100 ps and a few tens of nanoseconds, as achieved with a Q-switched laser. For large distances, high pulse energies are required. This can raise laser safety issues, particularly if the laser wavelength is not in the eye-safe region. For nanojoule to microjoule pulse energies (as required for moderate distances), it is possible to use a passively Q-switched microchip Er:Yb:glass laser, which can generate fairly short pulses (duration of the order of 1 ns) with pulse energies around 10 μJ in the eye-safe spectral region.
The RP Photonics Buyer's Guide contains 10 suppliers for time-of-flight measurement devices. Among them:
Questions and Comments from Users
Here you can submit questions and comments. As far as they get accepted by the author, they will appear above this paragraph together with the author’s answer. The author will decide on acceptance based on certain criteria. Essentially, the issue must be of sufficiently broad interest.
Please do not enter personal data here; we would otherwise delete it soon. (See also our privacy declaration.) If you wish to receive personal feedback or consultancy from the author, please contact him e.g. via e-mail.
By submitting the information, you give your consent to the potential publication of your inputs on our website according to our rules. (If you later retract your consent, we will delete those inputs.) As your inputs are first reviewed by the author, they may be published with some delay.