RP Photonics
Encyclopedia

Dr. Paschotta, the founder of RP Photonics, supports your R & D with his deep expertise. Save time and money with efficient support!

Powerful simulation software for fiber lasers and amplifiers, resonator design, pulse propagation and multilayer coating design.

The famous Encyclopedia of Laser Physics and Technology provides a wealth of high-quality scientific and technical information.

In the RP Photonics Buyer's Guide, you easily find suppliers for photo­nics products. As a supp­lier, you can profit from enhanced entries!

Learn on lasers and photonics every day!
 part of theVirtualLibrary

# Peak Intensity of Gaussian Beam

Posted on 2009-10-03 as a part of the Photonics Spotlight.

Ref.: encyclopedia articles on Gaussian beams and effective mode area

Isn't it easy: when a laser beam has the optical power P and the area A, its intensity is I = P / A. Not enough substance for an article in the Photonics Spotlight, you may think. I will demonstrate, however, that it is indeed worth to look at that, and you would be happy if others had done it.

The relation between intensity, power and beam area is trivial only for rectangular intensity profiles. More frequently, however, we are dealing with Gaussian beams. For arbitrary intensity profiles, one can define an effective mode area. One may then expect that the simple formula above remains valid when using that value for A. This is not true, however. For Gaussian beams, for example, the effective mode area is πw2 with the mode radius w, and the peak intensity (i.e., the intensity on the beam axis) is two times the power divided by the effective mode area. That factor 2 is often forgotten. Obviously, the same issue occurs in the context of the fluence (energy per area) of laser pulses.

Unfortunately, you cannot avoid the problem for yourself simply by always applying that factor. The trouble is that the scientific and technical literature is strongly contaminated with that error, and in most cases it is not even possible to find out whether or not that error has occurred.

As an example, consider specifications of an optical damage threshold, i.e., the maximum optical fluence which some device such as a dielectric mirror can tolerate. The usually measured parameters of the laser beam are the pulse energy and the mode radius. The peak intensity is not measured, but calculated from the other data. If those reporting results of their damage threshold measurements have not been aware of that factor 2, the quoted damage fluence is 2 times too low. (With a top-hat beam shape, for example, one could apply 2 times the energy to some area, compared with what would be expected from the wrong specification.) But if it is so, people are also not aware that such a detail has to be addressed, so you cannot even check whether or not that happened.

In my daily work, I encounter that problem quite frequently. Its consequence is that many literature values contain an uncertainty of a possibly missing factor 2. This is rather awkward in many cases. For example, when I have to design the laser resonator of a high-power Q-switched laser, I am not sure how large the mode areas on the mirrors have to be to avoid damage. Just staying far away from such limits is not always a solution, as this makes it more difficult to reach other design goals, such as a short resonator with low alignment sensitivity.

Of course, one can easily cause even more confusion, for example by specifying a "beam size", leaving it open whether a radius or a diameter is meant. That alone already makes for a factor of 4 in the intensities, if the specification is misunderstood.

Lay persons might be astonished to learn what level of sloppiness is common in science and technology. We would all profit from some more precision and care.

Note that you can also receive the articles in the form of a newsletter.

 Your general impression: don't know poor satisfactory good excellent Technical quality: don't know poor satisfactory good excellent Usefulness: don't know poor satisfactory good excellent Readability: don't know poor satisfactory good excellent Comments: Found any errors? Suggestions for improvements? Do you know a better web page on this topic? Spam protection: (enter the value of 5 + 8 in this field!)

If you want a response, you may leave your e-mail address in the comments field, or directly send an e-mail.

If you like our website, you may also want to get our newsletters!

# Thin-film Optics Software

RP Coating is a most versatile software for thin-film design. Analyze and optimize a wide range of devices:

Further features:

• extreme flexibility due to scripting: arbitrary parametrization e.g. of chirped layer structures, definition of arbitrary figure of merits for numerical optimization, generation of arbitrary plots
• comprehensive documentation (PDF and online)
• competent technical support

Use RP Coating to quickly become a thin-film expert!

– Show all banners –

– Get your own banner! –